Fiber Optic Tester Tool Optical Power Meter Optical Light Source 1310/1550 SM. 1 x Optical Power Meter -70+6 dbm, If you need -50+26dbm, Pls contact with me. 1 x Handheld Light SourceWavelength: 1310/1550nm , If you need other wavelength, pls contact with me. 2 x Portable Bag. 2 x FC Adapter. Our designs series optical power meter to meet the high demand. It intergrades the handheld optical power meter and the intelligent optical power meter in one unit. It can be used for the absolute power measurement and range of the link loss in dB. Its compact size, friendly operation interface, broad power measurement range, high precision and brand-new user automatic calibration function and high performance in application makes it an ideal tool for optical fiber network. Mode: Optical Power Meter. Wavelength Range(nm): 8001600. Measurement Range(dBm): -70+6. Accuracy: Linear 0.1% or Non. Auto-off Time(min): 10. Operating Time(h): 130. Battery: 3AA 1.5V (Not Include). Dimension(mm): 152 x 74 x 26. Wavelength Range: Specified standard operating wavelength range in which the Power Meter can work. Properly under certain technical specifications. Power Measurement Range: The maximum and minimum range in which the Power Meter can work properly. Uncertainty: Difference between two measurement results that were tested by Power Meter and another Standard Power Meter respectively. The LCD screen display the measurement tester in dB, dBm, mW, uW, nW unit; the selected wavelength; the current operating situation and so on. Press the key to turn the unit on/off. To test the power value under certain wavelength. Press the key for auto-zero. To switch the current operating wavelength between 850nm, 980nm, 1300nm, 1310nm, 1490nm, 1550nm. To turn the background light on/off. (1) Press the ON/Off key for a few secounds to turn on the unit. (2) Press the ON/Off key for a few secounds to turn off the unit. 2 Abdolute power measurement. (1) Turn on the power meter. (2) Press the key to switch between the wavelengths. The default wavelength is 1310nm. (3) Connect the light to be measured, and then reading will be displayed on the LCD screen, including linear and nonlinear value. 3 Relative power measurement. (1) Select the wavelength to be measured. (2) Under “Absolute power measurement mode”, connect to the light to be tested. (3) Press dB key, then current power value is stored as a a referance value in dB unit. At the same time it also display to current absolute power value and current relative is 0dB. (4) Connect to another beam of light to be tested, display the current relative power value and absolute power value under tested. It has 3 kinds of working mode. 4.1 Factory Mode: The factory is responsible for the test and calibration. 4.2 User Mode: Press the +Light Key synchronously, then enter the User mode. Numeral “1” will be displayed on the top right corner. Press the + Light Key synchronously once again, it exit from the user mode and enter the working mode. Number “0” will be displayed on the top right corner. Notes: If any errors or mistakes caused by the user self-calibration operation, please press the +Zero synchronously under the use mode to go back to the factory setting. (1) Auto-off: Press LIGHT + UNITS key to turn on the auto-off function. The auto-off symbol will be displayed on the top right. The unit will be turn off automatically after 10 minutes idle time. (2) Background light On/Off: Under the working mode, press LIGHT to turn the backpround light ON/Off. A little sun symbol will be displayed on the top right. Optical fiber loss measurement. Turn on optical power meter and press the key to selet the wavelength. Turn on optical light source(emitting source) and select the wavelength. Wait for 1-2 minutes until it stabilize. Select a piece of patch cords, which is used to connect with the light source, we call it emitting patch cords, Cleaning connector of the patch cords. The fiber of the emitting source patch cords must be the same fiber type as the fiber under test. Connect the light source (emitting source) with the handheld power meter through the emitting source patch cords. Handheld Power Meter get the power measurement value. This value should be close to the one that light source (emitting source) set. If it has wide disparity, please make sure the fiber connection is clean properly else replace another jumpers. Press the dB key; the reading 0.00dB will be displayed on the screen. The tested power values will be set to the reference value. Step 2-Optical Loss Measurement. Keep the emitting source patch cords connection with the light source (emitting source). Connect the Handheld Power Meter and light source (emitting source) to the optical fiber link respectively. Notes: cleaning all the connectors surface including all the necessary optical adaptors. The reading in dB unit displayed on the screen is the tested optical fiber link loss. (also display the current absolute optical power value in dBm unit). It is important to keep all optical connectors and surfaces free from oil, dirt or other contamination to ensure proper operation. Keep using one type of adapter to avoid excess loss from different connectors. Please use dust proof cap for protection to avoid begin scratched or contaminated when Handheld Power Meter not in operation. Light interface is sensitive, please carefully plug in and pull out the connectors. Please use clean cotton to clean the sensor surface, clean it in clockwise direction carefully. If does not need to use for a long time, please take out the battery. Handheld Light Source optical light source. Together with optical power meter, it acts as a perfect solution for the fiber optic network work. Ø Providing 14 wavelengths output and wavelengths can be optionalaccording to customers needs. Ø CW, 2Hz modulation output at 650nm, and CW, 270Hz, 1KHz, 2KHz. Modulation output at other wavelengths. Ø High stability of the output power. Ø Stable output wavelength. Ø Compact size and decent appearance. Ø Large LCD, easy operation. Wavelength(nm): 1310/1550If you need other wavelength, pls contact with me. Typical Output Power(dBm): -6+7dBm. Spectral Width(nm): 10. Modulation Frequencies: CW, 270Hz, 1KHz, 2KHz. Power Supply: 3AA 1.5V batteries (Not Include). Battery Operating Time(H): 45. Recommendation: Handheld light source is designed for optimal use with optical power meter for measuring optical loss on both single mode and multimode fiber cable. (1) Wavelengths output ports: FC/PC or other type connectors can be required. (2) Liquid Crystal Display: For wavelength display; Modulation indicator. (3) ON/OFF Key: Press to turn the Handheld Light Source Light On or Off. (4) Wave key: For wavelength selection. (5) Mode key: Modulation control: CW light output and modulated light output can be selected by pushing Mode key. When modulated light output is selected, a dot appears in LCD. Press ON/OFF key about one second to switch on Handheld Light Source, default wavelength is 1310nm. To use Handheld Light Source light source. Wavelength selection: Press Wave key for wavelength selection. Modulation: Press the MODE key, choose the CW, 270Hz, 1KHz or 2KHz light output. Switch off: Press ON/OFF key to switch Handheld Light Source off. If the Handheld Light Source is not used for a long time, the light output port must be protected with the dustproof hat. When the Handheld Light Source is used for a period of time, the output port maybe polluted. It results in the output power decreasing. So the connector must be cleaned with alcohol cotton. Once not using for a long time, please take out the battery. Dont take apart the Handheld Light Source please. DO NOT look into Handheld Light Source adapters directly when light source is on, laser output does harm to your eyes. The output light is unstable or Chage battery or re-charge battery. Decreases rapidly while operation. Output power is fluctuating. After warm-up for 15 minutes, then operate it. The output port is polluted. Clean the connector with alcohol smoothly. Or contact with supplier. If you want pay by T/T, Please ask for out Bank information. Please Check and Confirm the Part No. The item “Fiber Optic Tester Tool Optical Power Meter Optical Light Source 1310/1550 SM” is in sale since Wednesday, March 30, 2016. This item is in the category “Business & Industrial\Test, Measurement & Inspection\Test Meters & Detectors\Fiber Optic Power Meters”. The seller is “fiberoptictoolsshoppingmall” and is located in , . This item can be shipped to North, South, or Latin America, all countries in Europe, all countries in continental Asia, Australia.
- MPN: Does Not Apply
- Brand: Unbranded/Generic
- Type: Optical Light Source
- Model: Optical Power Meter
- Country//Region of Manufacture: China